Why is the Sequal Eclipse 3 the best all-round 24/7 portable oxygen concentrator?
The Sequal Eclipse is the ultimate device for portable oxygen. It is recommended for patients who wish to travel and require oxygen for more than 16 hours a day and use oxygen overnight. It can also be used in conjunction with CPAP and BIPAP, face masks, humidifer bottles and of course nose cannula!
The Sequal Eclipse 3 is highly recommended by Pure O2
This device can be used overnight on continuous oxygen mode between 0.5lpm and 3lpm. It can also provide up to 6lpm on pulse mode for good reliable ambulatory use, and is particularly effective with patients that struggle to maintain good O2 saturation levels.
Each battery can provide up to 5 hours’ usage at 1 to 4lpm in pulse mode, and the Eclipse can be run on DC car adaptor or any mains voltage system around the world. This means it is also fine to use on cruise ships, and is approved by most major airlines.
The Eclipse 3 has great new features that make it capable of supplying oxygen up to high level dosages of 6lpm (192ml per breath). It also monitors patient breathing every 15 seconds on pulse mode, and if it discovers that the patient is struggling to breathe, it automatically alternates between continuous flow and pulse to ensure that the patient receives oxygen correctly. For instance, if a patient falls asleep on the pulse mode the Eclipse will automatically switch to continuous flow to ensure maximum patient safety. Please call or email for pricing, as this product can only be supplied after consultation from one of our medical team!
~ Can be used overnight with CPAP and other Bi-Level devices in continuous mode
~ Auto SAT technology is a new technology that monitors patient breathing and guarantees a maximum dose for each breath
~ Can provide up to 192ml dosage of oxygen per breath, and in some cases can maintain patients requiring 8lpm
~ FAA approved for commercial air travel
~ Three pulse dose rise time levels on pulse mode - fast, medium and slow delivery per breath
~ 3-stage Sensitive mode for detection of breath on pulse mode
Updated manufacturer battery times May 2013
Each battery supplies in hours an average of:
1lpm pulse on demand = 5 2lpm pulse on demand = 4 ½
3lpm pulse on demand = 4 4lpm pulse on demand = 3 ½
5lpm pulse on demand = 3 6lpm pulse on demand = 2 ½
6lpm (128ml per breath) = 2 6lpm (160ml per breath) = 1 ¾
6lpm (190ml per breath) = 1 ½
0.5lpm continuous flow = 4 1lpm continuous flow = 3 ½
1.5lpm continuous flow = 2 ½ 2lpm continuous flow = 2
2.5lpm continuous flow = 1 ½ 3lpm continuous flow = 1 ¼
Features of the Portable Oxygen Concentrator Size:
H x W x D 19.3 x 12.3 x 7.1 inches / 49.0 x 31.2 x 18.0 cm
Weight: with Power Cartridge 17.4lb (7.9kg); without Power Cartridge 13.9lb (6.3kg)
Continuous flow settings 0.5 to 3.0lpm (0.5-litre increments)
Oxygen output pressure 5.0 psig (34.5 kPa) nominal pulse flow settings 1.0 to 6.0 (0.5 increments)
Trigger sensitivity adjustable between 0.25cm H2O to 0.7cm H2O
Sound level 48dB(A) at 3.0lpm continuous flow; 40dB(A) at 3.0 pulse flow setting
Full 3-year manufacturer’s warranty. Full after-sales service within the UK. Also, new 5-year warranties can be purchased with this model!
PLEASE NOTE: This product is only supplied to patients who can produce a valid doctor’s letter stating that they require oxygen. Failure to produce this may mean automatic cancellation of order!
VAT EXEMPTION: This product is supplied for medical uses only, and in most cases where patients or charities purchase or rent this equipment within the UK, VAT may not be charged or, if charged already, may be refunded in full. Please download the government information sheet and the VAT exemption form and fill in if applicable. If you have any questions please contact us via email on firstname.lastname@example.org, or call us in the UK on 0870 712 0202.
The Sequal Eclipse has changed my life for the better and the guys at PureO2 couldn't do enough to help. Thank you
Review by Chris (Posted on 02/03/2012)
We've just returned from a short break in Wales which gave us an excellent opportunity to road-test the Sequal Eclipse!
Bron had no problems at all using it and we were both pleasantly suprised by the quietness of the machine. Overall I'd say it was a great success.
Review by J.Bailey (Posted on 08/02/2012)